Bone Collector - Bone Collection Bur
Regular price $150.00
Couldn't load pickup availability
You can contact us for your orders and questions. +90543 456 2614
All your orders are sent by fast air cargo with UPS EXPRESS Shipping
Your orders will be prepared and shipped within 3 business days.
In order to facilitate the collection of autogenous bone in regenerative procedures, the triple blade drill is a great option. Its differentials are the presence of three blades that provide greater efficiency in the collection of autogenous bone, in addition to less vibration and greater precision during the collection procedure in the donor bed. The association of autogenous bone with bone substitutes enhances the clinical results of grafting procedures, due to the osteogenic and osteoinductive potential of the autogenous bone associated with the osteoconductive potential of the bone substitute. So make your autogenous bone harvesting procedure more predictable.
> Auto cutting power even at low rotation;
> Collects and particles the bone quickly, without loss of the collected material;
> The amount of bone collected can be equivalent to the size of the drill;
> Minimally invasive accesses: smaller flaps;
> Excellent postoperative period in the donor area;
> Can be used on any type of bone with excellent durability!
> Ease of cleaning;
Made of surgical stainless steel with DLC I ASTM F899 440 coating
> Length: 38 mm
> Diameter: 5.0 mm
> Fit: Contra angle
> Indicated rotation: 300 rpm with irrigation
> Durability: ~25 Procedures
The way of using the product and surgical techniques are inherent to the training of the professional.
Handling should be minimal and with care. Do not proceed with use without first checking the cleaning conditions of the product. Handle the product properly taking into account material limitations. Pay attention to the instructions for use before carrying out the necessary operations.
During the procedure, avoid tilting or prying, always use at 90° in the access region.
Controlled pressure .
Excessive pressure must be avoided at all times. High pressure can overheat and lead to thermal necrosis. In extreme cases, instrument failure cannot be excluded.
To avoid unwanted overheating, make sure there is sufficient irrigation. Undischarged material can lead to a decrease in its performance and lead to an increase in temperature.
> Up to 50 rpm without irrigation I Above 100 rpm with irrigation
Higher rotation = smaller granules
Smaller rotation = larger granules
The use of external irrigation (drip) during surgery will reduce the possibility of thermal necrosis.
Made of PEEK, it incorporates the advantages of extreme biocompatibility, excellent mechanical profile and withstand high temperatures.